for information on TestEquity Model 1055 (ICS 8099)? Click HERE >>
Model 1056 (ICS 9099) adds these new features over the Model 1055 (ICS
- 2x to 5x faster than the 8099
- Webserver supports F4T native commands
- New webserver "device control page" lets
you enter individual commands directly, great for debugging
- Supports Raw Socket connection
Model 1056 Ethernet Interface Converter is a specialized Ethernet to
Serial Interface that provides Modbus RTU packet communication to adapt the
F4 or F4T Controller used on TestEquity chambers to a network. The converter
lets you send simple ASCII read-write messages over the network to control
and query the chamber's controller. The converter does all of the Modbus
RTU packet formatting. An internal webserver
provides a web page to read and control the chamber's set points and event
outputs. The converter is packaged in a small minibox case. It can be used
with any TestEquity chamber by plugging the converter cable into the chamber's
serial interface connector. The TestEquity Model 1056 is based on an ICS
Electronics 9099 Ethernet-to-Modbus Interface which has been customized to
ensure compatibility with all models of TestEquity Chambers. It is 100% compatible
with programs that were written for TestEqutiy Model 1055 (ICS 8099) Ethernet-to-Modbus
An internal webserver provides web pages for setpoint
entry and readback, readback of actual chamber temperature and humidity, and
Event Output setting/readback. Webserver pages for the F4T controllers add
the ability to set manual ramp rate parameters.
Why use the Ethernet Converter when the F4T Controller has an Ethernet
The native F4T Ethernet Interface uses the Modbus TCP protocol. The 1056 Ethernet
Interface Converter converts the controller's RS-232 Modbus interface to
Ethernet. The converter takes care of the Modbus packet formatting so the command
structure is simplified. The most compelling reason why you would want to use
1056 Ethernet Interface Converter instead of the F4T's native Ethernet interface
is if you wanted to program using simple ASCII commands instead of Modbus
packets. Note that Watlow Composer software will only work with the native
F4T Ethernet interface.
The converter and chamber appear as a virtual instrument to the application
and are easily controlled by the different programming techniques that are
in common use with today's Test and Measurement systems.
If you program with LabVIEW, National Instruments' VISA library recognizes
VXI-11.3 instruments like the converter as a TCP/IP resource. NI's Measurement
and Automation Explorer finds and communicates with the converter as with any
other VXI-11.3 compliant device. If you include the converter in a LXI system,
the converter and its chamber will appear as an LXI compatible instrument and
can be found with the VXI-11 Discovery procedure.
If you program with VEE, Keysight's (Agilent) IO Library treats the converter
as any other LAN instrument and makes it available to your program. If you
are a Visual Basic or C/C++ programmer, you can write your program to make
calls to any VXI-11 compatible VISA library. You can also write your program
using a raw socket connection to the converter. If you use Linux or any flavor
of UNIX such as SunOS, IBM-AIX, HP-UX, or Apple's OS X, you can communicate
with the 9099 through RPC over TCP/IP.
The user sends commands to the converter that sets the Modbus device address,
specifies the register to be read or written to and the data value. The converter
converts these commands into the Modbus packet format, adds the CRC checksum
and transmits the messages to the F4, F4T, or EZ-Zone controller in the chamber.
Received packets are checked and the responses to queries are returned to the
network client. The converter command syntax includes an optional question
mark to enable automatic response reading by included VXI-11 Keyboard Control
program and other smart programs. Modbus communication faults, exception messages
and other errors are reported to the user through a Modbus error register in
the converter's 488.2 Status Structure. The user can set up the 8099's IEEE-488.2
Status Reporting Structure to generate a Service Request (similar to a GPIB
SRQ) on an error or simply read the Modbus Error register if a problem occurs.
VXI-11 is a Ethernet instrument control protocol developed by the VISA
committee when they established the VXI and VISA standards. VXI-11 provides
GPIB like control of Ethernet instruments using RPC on TCP/IP. VXI-11 is a
packet protocol designed to give the user error free communication with the
instrument. Every VXI-11 command receives a response packet with command status
information and the appropriate response data. The converter has an expanded
IEEE-488.2 Status Reporting Structure that includes a register for reporting
Modbus communication faults. The converter's Status Reporting Structure can
be used to generate Service Requests using a Reverse Interrupt channel when
an error occurs Alternately, the user can periodically query the Modbus Error
and other register to check for command and communication errors.
Raw Socket Operation
The converter accepts telnet compatible, raw socket communication so that a
non-VXI-11 user can send simple read-write messages over the network to control
and query the chamber controller. This makes it easy to communicate
with the converter without a VISA library. Reverse channel notification of
an error is not possible with the raw socket protocol..
NOTE: TestEquity 1055 Ethernet Interface Converter is NOT compatible
with Watlow's WatView or Composer software.
Fully VXI-11.3 compliant
VXI-11.3: Device Interface
Sockets: 15 + 1 for UDP
Channel types: Data, Abort and Interrupt
Interface Names: intr0 for general use
Conforms to ONC RPC Version 2, VXI-11
Type: IEEE 802.3 compliant, Auto MDIX
Speeds: 10BaseT (10 Mb/s), 100BaseT (100 Mb/s)
IP Address: Static or DHCP with fallback to an AutoIP
Factory setting: 192.168.0.254 Static IP
Interface name: any [inst0]
Provides the following HTML 4.01 compatible web pages: Welcome, Configuration,
Confirmation, Reboot, Control.
Preloaded TestEquity-specific web pages included for temperature and temperature-humidity
chambers. Functions include Setpoint entry and readback, readback of actual
chamber temperature and humidity, and Event Output setting/readback. Webserver
pages for the F4T controllers add the ability to set manual ramp rate parameters.
IEEE 488.2 Capabilities
Runs all required 488.2 Common Commands, incorporates an extended IEEE-488.2
Status Reporting Structure and the Message Exchange Protocol.
The converter conforms to the SCPI 1994.0 Specification.
Full duplex serial interface with single ended RS-232 and differential
RS-422 (RS-485) signals. Signal selection made by jumpers on the converter.
RS-485 half-duplex operation enabled with a SCPI command.
RS-232 Signals: TxD, RxD, RTS, CTS, DSR and DTR
RS-422 Signals: Tx and Rx pairs
Baud Rates: 300, 600, 1.2K, 2.4K, 4.8K, 9.6K, 19.2K and 38.4K baud (Note: will
be configured for 9600 baud to be compatible with controllers used on TestEquity
Data Bits: 7 or 8 bits
Parity Stop Bits: Odd, even or none 1 or 2
Modbus RTU Commands
Modbus commands accept ASCII decimal values or HEX values starting with #h.
Code is the Modbus RTU command code produced by the 9099. Integer and register
values from 0 to 65,535. Floating Point per IEEE-754.
PWR - Indicates power on
LAN - Unit connected to an active LAN
ACT - Transferring messages to/from the network
RDY - Unit has passed self test TALK Unit asked to send data
LSTN - Unit sent a command or data
SRQ - Device Service Request asserted
ERR - Blinks for a soft VXI-11 error and On for command errors
Power Front panel switch
LAN Rear panel push-button that resets Reset the network settings.
Size, L x W x H : 7.29 x 1.52 x 7.45 inches (1185.2 x 38.6 x 189.2 mm)
Weight: 3 lbs. (1.4 kg.)
Construction: Lead Free
Serial: DB9 male connector with lock studs
Operation -10° C to +70° C
Storage -20° C to +85° C
0-90% RH without condensation
9 to 32 Vdc @ 3 VA
LAN Crossover Cable
9-Pin to 9-Pin Serial null-modem cable for connection to TestEquity chamber
Support CD-ROM with VXI-11 Utilities.
UL/CSA/VDE approved universal AC power Adapter